Otttobock Premium Telescoping Post-Op Knee Brace with buckle closures and silicone grip foam and malleolus suspension ensure the brace maintains proper positioning for healing. Range of Motion Knee brace limits motion between 0 degree to 120 degree and locks between 0 degree and 45 degree. Telescoping Uprights adjust from 21.5-inch to 28-inch allowing for optimal suspension from the malleolus and cuffs avoid incision sites for individualized fitting and increased patient comfort.
Otttobock Premium Telescoping Post-Op Knee Brace Features
Premium full foam design
Packaged for quick and easy application in an OR or recovery room setting
Buckle closures ease donning and doffing
Indications: - For locked or limited range of motion control following injury or operative procedures of the knee
Note: Measure leg circumferences 6" (15.25 cm) above mid-patella
More On Otttobock Premium Telescoping Post-Op Knee Brace
Align the brace with the leg making sure the center of the hinge is aligned with the knee cap.
Adjust the length using the push button telescoping uprights.
Adjust the range of motion according to the physicians directions. Press the flexion button and slide the button to the prescribed setting.
Press the extension button and slide the button to the prescribed setting.
Ensure both hinges are set to the same prescribed settings. Note: You may need to disengage the drop lock in order to adjust the hinge range of motion.
Open the brace by disengaging the buckle closures.
With the patient sitting or lying down, slide the brace under the affected leg.
Ensuring the hinges and uprights are aligned with the midline of the leg, remove the strap slack in the back of the brace.
Secure the buckle and adjust the strap tension using the hook and loop closure. Do not over tighten.
Trim if needed.
Repeat for the remaining straps.
Cleaning and Care:
Remove softgoods, hinges and uprights can be wiped clean with a damp cloth. Soft goods can be hand washed with mild detergent. Triple rinse, squeeze and air dry. Avoid twisting the material to remove water.
This device does not prevent injury and is not intended to reduce or eliminate the risk of injury.
Always consult with your physician or healthcare provider before making changes to the brace.
Carefully read all instructions for use before using or fitting this product.
Proper rehabilitation and activity modification are also a part of a safe treatment program.
If you experience pain, swelling, or any unusual reaction, contact your physician or healthcare provider immediately.
This device is offered under order of a physician or other qualified healthcare provider.
This device is intended for single patient use and is not intended to be reused on a second patient.
Contact customer service at 877-213-2018 or firstname.lastname@example.org to obtain a Return Authorization Number (R.A. #). This is valid for 30 days.
Please follow the instructions below to complete the process:
Display your RA number prominently on the outside of the packing box. Your RA number can also be found in your Returns list.
All returns and paperwork should be sent to the following address:
Ottobock Healthcare Attn: Returns Department 3820 W. Great Lakes Salt Lake City, UT 84120-7205
Please Note: Issuing of an R.A. is not a guarantee to provide credit or replacement. Decisions are made only after product is evaluated. For more details, download and review our terms of sale, return policy, and warranty information. Download 0.4 MB | PDF
Ground orders must be placed by 4:00 p.m. Central Time to ship same day. Next or 2nd Day Air orders must be placed by 6:00 p.m. Central Time to ship same day. (Excludes weekends and holidays)